I have lost or damaged one of the front panel signal connectors on my Tektronix BERT scope. How do I obtain a replacement connector?
A replacement connector is Tektronix p/n 131906200. These connectors are designed to be easily replaceable as they unscrew from the permant front panel input connector. When damaged, you remove the damaged connector and replace with a new connector.
Image of connector went removed from instrument:
This connector and base connector style is part of the PLANAR CROWN ® Universal Connector System.
產品系列： BERTScope BSA 系列
常見問答集 ID 64646檢視所有常見問答集 »