I would like to measure accurately the length for a 10 ft cable with the TDR function. I have used an 11803 before which had a set cursors to zero function which I cannot find on the TDS8000?
The TDS8000 does not provide the set cursor to zero function. To make the measurements with a TDS8000 and the cursors use magnification and magnify each end of the cable. The scope now displays Mag 1 and Mag 2, assign the cursors to each Mag and read the delta. For better accuracy use the delay measurement feature and assign the Delay measurement to Mag 1 and Mag 2. The TDS8000 has 3 time basses so there is no loss in resolution when using multi magnification windows..
常見問答集 ID 61911檢視所有常見問答集 »